Standard Interpretations - (Archived) Table of Contents|
MEMORANDUM FOR: JACKIE ROGERS Office of Health Compliance Assistance THRU: STEVE MALLINGER Deputy Directorate of Technical Support FROM: Pat Bertsche Directorate of Technical Support SUBJECT: 1. Life-GARD needle shield by Willander 2. The Recapper by Atomic Products CorporationI have been sent two syringe recapping devices by OSHA field personnel to render a professional opinion as to the effectiveness of the devices. As you well know, OSHA Instruction CPL 2-2.44A, subject: Enforcement Procedures for Occupational Exposure to HBV and HIV, states that needles shall not be recapped by hand.
I consulted with personnel in the Directorate of Health Standards Program who are currently working on the development of the bloodborne pathogens standard regarding these two devices.
It is my opinion, and the opinion of Health Standards personnel, that the Life-GARD needle recapping device will not provide adequate protection since the shield is not big enough to cover the entire hand, especially a large hand.
The Recapper syringe recapping device appears to provide adequate protection
for larger gauge needles. I unsuccessfully tried recapping a 1cc 25G 5/8
tuberculin syringe and needle with the Recapper. Therefore, the device
should only be used with larger gauge needles.
|Standard Interpretations - (Archived) Table of Contents|