Standard Interpretations - Table of Contents|
| Standard Number:||1910.1043; 1910.95|
December 14, 1984
Robert W. Patterson, M.D.
Family Medical Center
Post Office Box 1860
Sanford, North Carolina 27330
Dear Dr. Patterson:
This is in response to your letter dated November 5 regarding the Occupational Safety and Health Administration's (OSHA) regulations on the regular calibration of equipment, particularly spirometers and audiometers.
The best reference to answer your questions is OSHA's General Industry standards (enclosed).
Specifically, with regard to spirometers, minimum standards can be found in Appendix D of section 1910.1043 -- Cotton Dust standard (copy enclosed). There is no reference to regular calibration but rather the outlined standards must be met each time spirometry is done. Appendix C of this standard contains the predicted values for the FEV-1 and the FVC for males and females. A correction factor of 0.85 is used for blacks to adjust for physiological differences.
Calibration requirements for audiometers can be found in paragraph (h)(5) of section 1910.95 -- Noise standard (copy enclosed). Specifically, the functional operation of the audiometer shall be checked before each day's use, an acoustical calibration shall be done at least annually in accordance with Appendix E of the standard, and finally an exhaustive calibration should be done every two years in accordance with the American National Standard Specifications for Audiometers, S3.6-1969. Specifications for Audiometers can be obtained by writing the American National Standards Institute, Inc., 1430 Broadway, New York, New York 10018.
We hope this information answers your questions. Please let us know if we can be of further assistance.
Edward J. Baier
Directorate of Technical Support
November 5, 1984
Occupational Safety & Health
200 Constitution Avenue, NW
Washington, D.C. 20201
ATTN: ASSISTANT SECRETARY OF LABOR
I am very interested in obtaining information as to the OSHA regulations for regular calibrations of equipment particularly spirometers and audiometers. We are interested in doing industrial medicine work, however, wanted to get the guidelines prior to beginning this. I look forward to your response.
Robert W. Patterson, M.D.
|Standard Interpretations - Table of Contents|
The Department of Labor does not endorse, takes no responsibility for, and exercises no control over the linked organization or its views, or contents, nor does it vouch for the accuracy or accessibility of the information contained on the destination server. The Department of Labor also cannot authorize the use of copyrighted materials contained in linked Web sites. Users must request such authorization from the sponsor of the linked Web site. Thank you for visiting our site. Please click the button below to continue.