OSHA IMIS Code Number: Q118
Primary Laboratory Sampling/Analytical Method (SLC1):
** All Trademarks are the property of their respective owners.
- sampling media: Coated XAD-2 Tube (150/75 mg sections, 20/60 mesh); coating is 10% (w/w) 2-(Hydroxymethyl)piperidine
analytical solvent: Toluene
maximum volume: 5 Liters
maximum flow rate: 0.05 L/min
current analytical method: Gas Chromatography; GC/NPD with GC/MS identification
method reference: OSHA Modified NIOSH Analytical Method (NIOSH 2539 [255 KB PDF, 10 pages])
method classification: Partially Validated
note: OSHA modification is use of GC/NPD in place of GC/FID. This is a screening method and is not quantitative.
Accessibility Assistance: Contact the OSHA Directorate of Technical Support and Emergency Management at (202) 693-2300 for assistance accessing PDF materials.